Item #: 1145 |
Categories: Data Acquisition - WE7000
What is the Sampling Interval of the WE7241 Digital Thermometer Module?
The sampling interval changes with the type of time base and the number of measurement channels per module. There are two possible methods:
- If Time base = internal
Number of channels : 1-2 = 0.2 s, 3-5 = 0.3 s, 6-7 = 0.4 s, 8-10 = 0.5 s. (When modules are connected in a series, the sampling interval is that of the slowest module in the series.)
- If Time base = external
2 s or more (regardless of the number of channels).