Item #: 1230 |
Categories: Power Meter - WT3000
Why Won't the Measured Values of the Fundamental Wave Components and the Bar Graph of the Harmonics Display on the WT3000?
For harmonic measurement of a 50 Hz distorted wave signal, the fundamental frequency is 50 Hz and the width of the analysis window is ten waves, so the data measurement interval is 200 ms. Since (data measurement interval + data computation interval) = approximately 300 ms or more, select a data update rate of 500 ms or more.