Item #: 1175 |
Categories: Data Acquisition - WE7000
Why Does the Self Test Fail on the WE7111 Digital Oscilloscope Module?
You must use a probe with a attenuator of "10:1", otherwise the self test will not run correctly. Also, before executing the self-test, please allow for a 30 minutes warm-up time. If the instrument does not warm up, the self test may not be accurate.