Item #: 1149 |
Categories: Data Acquisition - WE7000
What is the Update Cycle of the D/A Output of the WE7141 100Mhz Universal Counter Module?
The update cycle changes depending on the acquisition mode and the gate time during frequency measurement. When the acquisition mode is One Shot, the update cycle is approximately 0.5-1 seconds. However for frequency measurement, it's about 10 seconds when the gate time is 10 seconds. When the method is Free Run, the update cycle occurs at each sampling interval. However for frequency measurement, it occurs approximately at each (gate time + sampling interval).