To change the voltage range on element 1 to 30 V, send the "CONFIGURE:VOLTAGE:RANGE 30V" command to the measuring instrument.
To change the current range on element 1 to the 1 A range, send the "CONFIGURE:CURRENT:RANGE 1A" command to the measuring instrument.
After entering settings, confirm the settings by sending the following commands:
For information on the voltage and current values that can be set using these commands, see the following:
Yokogawa, the world's largest manufacturer of energy and power analyzers and meters, provides a broad choice of digital power analyzers to satisfy all requirements.