If the only active module in the DL850 Chassis is the 720220 16 Channel Voltage Module, the waveform data file (ASCII or WDF) will NOT contain repeated samples. If multiple modules, including the 720220, are installed in the DL850 chassis, the waveform data for the 720220 WILL contain repeating data. The channels on the 720220 module will still MUX. If all 16 channels are active and the sample rate is set to 200kSa/sec, then each of the sub-channels will run at 5kSa/sec.
When saving the waveform data to an ASCII or WDF file format, if you wish to prevent the 720220 module from repeating samples, do the following:
Please download and view the files provided, for examples of repeating and non-repeating sample data.
Yokogawa data acquisition systems give you the most flexibility and power to measure, display, store, and even actuate any number of physical or electrical phenomena.