Yokogawa clamp-on current meter/multimeter 96061
φ18 mm AC 50 A, Load current measurement
Read current clamp probe with oscilloscope
1. Connect the output connector to the Input terminal of the measuring instrument.
2. Press the trigger to open the transformer jaws and clamp onto one conductor.
The measured conductor shall be at the center of the jaws.
When connecting a sensor with a power meter, match the arrow mark (power source to
load), which is indicated on the transformer jaws, with the current flowing direction in order to
synchronize the phases of measured current and output voltage.
3. Ensure that the tips of transformer jaws are firmly closed.
Looking for more information on our people, technology and solutions?