To change the voltage range on element 1 to 30 V, send the "INPUT:VOLTAGE:RANGE:ELEMENT1 30V" command to the measuring instrument.
To change the current range on element 1 to the 1 A range, send the "INPUT:CURRENT:RANGE:ELEMENT1 1A" command to the measuring instrument.
After entering settings, confirm the settings by sending the following commands:
To set/query the ranges on all elements at once, use the following commands:
For information on the voltage and current values that can be set using these commands, please refer to:
The PZ4000 is designed for High Frequency Power Measurements as well as Transient Power Measurements. Its real-time acquisition system can perform cycle-by-cycle power measurements. One to Four input elements offer Basic Power Accuracy of 0.2% with 2 MHz Bandwidth and 5MS/sec digitizing rates.
The High-end WT1600 is designed for a wide range of applications, from energy-saving to large load applications. The WT1600 has a six element input capability to provide maximum flexibility, and Basic Power Accuracy of 0.1% with 1 MHz Bandwidth.
Yokogawa, the world's largest manufacturer of energy and power analyzers and meters, provides a broad choice of digital power analyzers to satisfy all requirements.