The 720931 passive clamp-on current probe enables current waveform readings directly via analog input modules on ScopeCorders including the DL350 and DL850 using isolated BNC connections.
1. Connect the output connector to the Input terminal of the measuring instrument.
2. Press the trigger to open the transformer jaws and clamp onto one conductor. The measured conductor shall be at the center of the jaws. When connecting a sensor with a power meter, match the arrow mark (power source to load), which is indicated on the transformer jaws, with the current flowing direction in order to synchronize the phases of measured current and output voltage.
3. Ensure that the tips of transformer jaws are firmly closed.
|Measuring Range||AC 0 to 200 A RMS (300 A Peak)|
|Accuracy (sine wave input)||Frequency|
|50/60 Hz||±0.5% of Reading ±0.1 mV|
|40 Hz to 1 kHz||±0.8% of Reading ±0.2 mV|
|1 kHz to 3.5 kHz||±1.0% of Reading ±0.4 mV|
|Maximum Allowable Input||AC 250 A RMS Continuous (50/60 Hz)|
|Output Voltage||2.5 mV/A||AC 0 to 500 mV RMS|
|Maximum Conductor Size||30mm Ø|
Please download the manual for complete specifications.
Procurando mais informações sobre nossas pessoas, tecnologia e soluções?