In free run mode, the execution timing is different depending on the type of computation. In the case of computations such as FFT, Filter, pulse width, etc. various data are arranged in a chronological direction. Calculations cannot be carried out during measurement. Once the measurement has been completed, the computation is performed and the results are displayed on the waveform monitor.
For computations using the four mathematical operations, exponents, logarithms, and trigonometry, you can perform them anytime during measurement (even if there was only one data available for calculation) and display the results on the waveform monitor. Please see the Computation Waveform Viewer Instruction Manual IM 707714-61E.
One system, multiple instruments: WE7000 satisfies demands for fast, reliable and precise data acquisition which uses a standard laptop or PC as its user interface. Input modules plug into an expandable measuring station. Via the optical interface, it is possible to galvanically isolate the laptop/PC.
Yokogawa data acquisition systems give you the most flexibility and power to measure, display, store, and even actuate any number of physical or electrical phenomena.