The correct syntax for implementing the Atan function on the DL850 is ATAN(C1).There is a typo in the DL850 IMDL85001EN 4thEd Features Guide, ATAN(C1,C2) is in-correct. However, the syntax for the DL750 __IS__ ATAN(C1,C2).

The Atan function on the DL850 calculates the arc tangent of the waveforms from Source 1 and Source 2, and outputs an angle in radians within ±π. This differs from typical Atan functions, which calulates the inverse tangent of a number. The formula for Atan calculation on the DL850 is:

Angle = Atan(s1/s2)

where s1 and s2 is the equivalent sampled data from Source 1 and Source 2.

Angle = Atan(s1/s2)

where s1 and s2 is the equivalent sampled data from Source 1 and Source 2.

Data acquisition (DAQ) is the measurement, recording, analyzing, and presentation of real world phenomena. It includes electrical measurements such as voltage, current, power as well as measurements through sensors and transducers including temperature, pressure, strain, vibration, and more.

Yokogawa high speed data acquisition systems deliver industry leading isolation, bit resolution, sampling rate, and memory depth, with independent channel hardware and easy to use software.

Capture, analyze, and troubleshoot electrical or real-world physical signals. Oscilloscopes observe the change of electrical signals over time, continuously graphed on a display as voltage or amplitude vs. time.

A ScopeCorder is an instrument combining a mixed signal oscilloscope and portable data acquisition recorder into a modular platform designed to capture both high-speed transients and low speed trends. Yokogawa’s ScopeCorder product family provides flexible and high-performance multi-channel test instruments by combining a variety of signal conditioning input modules, on-board calculations, and deep data acquisition storage into a off-the-shelf data acquisition solution.

Top