The DL850 can save measured waveform data to a file in Binary, ASCII, or Floating Point format. You can use the FILE Waveform
Save Menu or the SAVE MENU menu to set how files are saved.
To save using the SAVE MENU method:
To save using the FILE Waveform Save method:
The extension for each Data Type is:
Note that in Dual Capture mode, waveforms saved in ASCII .CSV format are saved simultaneously into two files labeled *.CSV and *DC.CSV. The main waveform data is stored in the first file while the captured waveform data is saved in the second file, with DC added to the end of the file name.
Saving the waveform data in Binary .WDF and converting it to .CSV using Xviewer will not work, as doing so will only save the main waveform data. All other captured waveform data will be subsequently lost.
Yokogawa data acquisition systems give you the most flexibility and power to measure, display, store, and even actuate any number of physical or electrical phenomena.
Yokogawa high speed data acquisition systems deliver industry leading isolation, bit resolution, sampling rate, and memory depth, with independent channel hardware and easy to use software.
World-class digital oscilloscopes from Yokogawa: The digital oscilloscopes have high-speed sampling and a wide range of bandwidths that can be utilized for design and development of electronic devices. The ScopeCorders have the advantages of both a digital oscilloscope and a multi-channel data recorder.
A ScopeCorder is an instrument combining a mixed signal oscilloscope and portable data acquisition recorder into a modular platform designed to capture both high-speed transients and low-speed trends. Yokogawa’s ScopeCorder product family provides flexible and high-performance multi-channel test instruments by combining a variety of signal conditioning input modules, onboard calculations, and deep data acquisition storage into an off-the-shelf data acquisition solution.