The DL850 can save measured waveform data to a file in Binary, ASCII, or Floating Point format. You can use the FILE Waveform
Save Menu or the SAVE MENU menu to set how files are saved.
To save using the SAVE MENU method:
To save using the FILE Waveform Save method:
The extension for each Data Type is:
Note that in Dual Capture mode, waveforms saved in ASCII .CSV format are saved simultaneously into two files labeled *.CSV and *DC.CSV. The main waveform data is stored in the first file while the captured waveform data is saved in the second file, with DC added to the end of the file name.
Saving the waveform data in Binary .WDF and converting it to .CSV using Xviewer will not work, as doing so will only save the main waveform data. All other captured waveform data will be subsequently lost.