The DL850 can save measured waveform data to a file in Binary, ASCII, or Floating Point format. You can use the FILE Waveform
Save Menu or the SAVE MENU menu to set how files are saved.
To save using the SAVE MENU method:
To save using the FILE Waveform Save method:
The extension for each Data Type is:
Note that in Dual Capture mode, waveforms saved in ASCII .CSV format are saved simultaneously into two files labeled *.CSV and *DC.CSV. The main waveform data is stored in the first file while the captured waveform data is saved in the second file, with DC added to the end of the file name.
Saving the waveform data in Binary .WDF and converting it to .CSV using Xviewer will not work, as doing so will only save the main waveform data. All other captured waveform data will be subsequently lost.
Data acquisition (DAQ) is the measurement, recording, analyzing, and presentation of real world phenomena. It includes electrical measurements such as voltage, current, power as well as measurements through sensors and transducers including temperature, pressure, strain, vibration, and more.
Yokogawa high speed data acquisition systems deliver industry leading isolation, bit resolution, sampling rate, and memory depth, with independent channel hardware and easy to use software.
Capture, analyze, and troubleshoot electrical or real-world physical signals. Oscilloscopes observe the change of electrical signals over time, continuously graphed on a display as voltage or amplitude vs. time.
A ScopeCorder is an instrument combining a mixed signal oscilloscope and portable data acquisition recorder into a modular platform designed to capture both high-speed transients and low-speed trends. Yokogawa’s ScopeCorder product family provides flexible and high-performance multi-channel test instruments by combining a variety of signal conditioning input modules, onboard calculations, and deep data acquisition storage into an off-the-shelf data acquisition solution.