Model 2793 resistance box is high-accuracy, stable DC variable resistor with 6 dials and is available in two styles: 279301 for medium resistance from 0.1 to 1,111.210 Ω in 1 mΩ steps (best suited for calibration of resistance thermometers or bridges); 279303 for high resistance from 0 to 111.1110 MΩ in 100 Ω steps (suitable for calibration of insulation resistance testers or bridges).
Dielectric Strength | 1,000 V AC for one minute between Panel and Circuit |
---|---|
Insulation Resistance | More than 500MΩ at 500V DC between panel and circuit |
Dimension | 497（W）×116（H）×140（D）mm, excluding protrusions 19.57 x 4.57 x 5.51 in. |
Weight | Approx. 4.8kg 10.6 lbs |
Resistance Range | 0.1mΩ to 1,111.2100Ω (minimum resistance is 0.1Ω) |
Dial Composition | 0.001Ω x 10 + 0.01Ω x 10 + 0.1Ω x 11 + 1Ω x 10 + 10Ω x 10 + 100Ω x 10 |
Resolution | 1mΩ |
Accuracy | ±(0.01Ω + 2mΩ) at temperature of 23 ±2°C, humidity of 45 to 85%, and 0.1W Power Application |
Max. Allowable Input Power | 0.2SW/slep. Within 1W for overall instrument. |
Max. Allowable Input Current | 50mA (100Ω steps), 150mA (10Ω steps), 500mA (10 Ω steps) and 1.5A (0.1Ω steps) |