Description | ACA, DCA, ACV, DCV, OHM, Freq. °C Continuity Check |
---|---|
Method of detection | Mean Value |
Display | LCD (Digital display: 3999 counts) |
DC Current Range | 400/2000A |
DC Current Resolution | 0.1A |
Response Time | Approx. 2 seconds |
Range Switching | Auto-range (on resistance range) Manual-range (on current, voltage range) |
Operating temperature and humidity | 0 - 40°C, 85% RH or less (no condensation) |
Influence of external magnetic field | 4A or less at 400 A/m |
Influence of conductor position | ±(1.5% rdg + 3 dgt) or less |
Safety Standard | Conforms EN61010-1, EN61010-2-031, EN61010-2-032 |
Circuit voltage | 1000 Vrms or less |
Withstanding voltage | 5.55 kV AC for one minute |
Power Supply | LR6 x 2 R6P (SUM-3) x 2 |
Battery Life | Approx. 100 hours (continuous) |
Consumed current | Approx. 9 mA |
Sleep Function | Automatically powered down in about 10 minutes after the last switch operation |
Diameter of measurable conductor | 55 mm at maximum |
Dimensions | Approx. 105 (W) x 250 (H) x 49 (D) mm |
Weight | Approx. 530 g |
Accessories | Batteries, carrying case (93034) Output plug (98012) Test Lead (98011) User's manual (IM CL250) |
Peak Hold | On all range |
Max Hold | On current / Voltage range |
AC Current Range | 400/2000A |
---|---|
AC Current Resolution | 0.1A |
AC Current Frequency characteristics | 40~1kHz |
AC Current Method of detection | Mean Value |
AC Voltage Measurement | O |
DC Voltage Measurement | O |
Continuity Check Measurement | O |
Frequency Measurement | - |
Temperature Measurement | - |
Recorder output Function | O |
Waveform monitor output Function | - |
Data hold Function | O |
Auto hold Function | - |
Peak Hold Function | - |
Maximum value memory Function | O |
Mean value display Function | - |
Filter Function | - |
Range Hold Function | (current range) |
Auto ZERO Adjust Function | O |
Auto power-off Function | Δ |
At 23°C±5°C，75%RH or less
Accuracy: ±( % rdg + dgt) = ±(% readout + value of least significant digit)
Parameter | Reference | Accuracy |
---|---|---|
DC current | 400/2000A | 1.5+2 |
AC current | 400A/2000A(0～1000A) | 1.5+2 (50/60Hz) |
3.0+4 (40～500Hz) | ||
5.0+4 (500～1kHz) | ||
2000A (1001～2000A) | 3.0+2 (50/60Hz) | |
DC voltage | 400/1000V | 1.0+2 |
AC voltage | 400/750V | 1.5+2 (50/60Hz) |
1.5+4 (40～1kHz) | ||
Resistance | 400/4000Ω | 1.5+2 , Beeps at below 50±35Ω (continuity check) |
DC output | DC400A (0～400mV) | ±1.5% rdg ±3mV |
DC2000A (0～200mV) | ±1.5% rdg ±3mV | |
AC400A (0～400mV) AC2000A (0～100mV/0～1000A) | ±1.5% rdg ±3mV (50/60Hz) | |
±3.0% rdg ±3mV (40～500Hz) | ||
±5.0% rdg ±3mV (500～1kHz) | ||
AC2000A (100.1～200mV/1001～2000A) | ±3.0% rdg ±3mV (50/60Hz) |