When viewing the ASCII Header File *.HDR, each timestamp row corresponds to the time a trigger occurs. Multiple timestamp rows will be created if the trigger mode is set to Single(N) and History is set to "All." The DL750 will save the trigger times of N waveforms into separate timestamp rows in the ASCII header file.
If the record length is set short on the DL750, the memory is divided into blocks, and waveform data is saved into individual blocks each time a trigger occurs. In sequential store function, Single(N) trigger mode, you can specify the waveform acquisition count (block number), and the measurement stops when the specified count is reached.
Please download the sample HDR file, and open it using Notepad for an example of multiple timestamp rows.
Yokogawa data acquisition systems give you the most flexibility and power to measure, display, store, and even actuate any number of physical or electrical phenomena.
Yokogawa high speed data acquisition systems deliver industry leading isolation, bit resolution, sampling rate, and memory depth, with independent channel hardware and easy to use software.
World-class digital oscilloscopes from Yokogawa: The digital oscilloscopes have high-speed sampling and a wide range of bandwidths that can be utilized for design and development of electronic devices. The ScopeCorders have the advantages of both a digital oscilloscope and a multi-channel data recorder.
A ScopeCorder is an instrument combining a mixed signal oscilloscope and portable data acquisition recorder into a modular platform designed to capture both high-speed transients and low-speed trends. Yokogawa’s ScopeCorder product family provides flexible and high-performance multi-channel test instruments by combining a variety of signal conditioning input modules, onboard calculations, and deep data acquisition storage into an off-the-shelf data acquisition solution.