For a strain gage factor greater than 2.20, please set the strain gage factor to 2.0 and use linear scaling. Refer to the example below.

For X > 2.20:

- set gage factor of strain module to 2.00

- set linear scaling constant A equal to a value of 2.0/X.

For example, if X equals to 2.21, set scaling constant A to 2.00/2.21 or 0.905.

- set gage factor of strain module to 2.00

- set linear scaling constant A equal to a value of 2.0/X.

For example, if X equals to 2.21, set scaling constant A to 2.00/2.21 or 0.905.

Related Information:

This method is also valid for the WE7245 module of the WE7000 series, for a gage factor greater than 2.30.

This method is also valid for the WE7245 module of the WE7000 series, for a gage factor greater than 2.30.

Data acquisition (DAQ) is the measurement, recording, analyzing, and presentation of real world phenomena. It includes electrical measurements such as voltage, current, power as well as measurements through sensors and transducers including temperature, pressure, strain, vibration, and more.

Yokogawa high speed data acquisition systems deliver industry leading isolation, bit resolution, sampling rate, and memory depth, with independent channel hardware and easy to use software.

Capture, analyze, and troubleshoot electrical or real-world physical signals. Oscilloscopes observe the change of electrical signals over time, continuously graphed on a display as voltage or amplitude vs. time.

A ScopeCorder is an instrument combining a mixed signal oscilloscope and portable data acquisition recorder into a modular platform designed to capture both high-speed transients and low speed trends. Yokogawa’s ScopeCorder product family provides flexible and high-performance multi-channel test instruments by combining a variety of signal conditioning input modules, on-board calculations, and deep data acquisition storage into a off-the-shelf data acquisition solution.

Top