The software allows you to enter the voltage and current range, and other settings in the WT series instrument's display panel. Also you can display all measurement item data (normal and calculated) on a PC, and save it to an ASCII file.
Note: For the WT110/ WT130, these features are only available with firmware version 2.01 or later.
PCA PC with a 80486DX2 CPU or later (Pentium or later recommended), and 8 MB or more of memory.
Windows 98, Windows Me, Windows 2000 Professional, or Windows XP
Floppy disk drive
3.5" floppy disk drive using 1.44 MB (MS-DOS compatible) type disks
A hard disk drive with 10 MB of available free space
A video adapter capable of displaying at 640 x 480 or higher resolution (1024 x 768, 16 bit color or higher recommended)
VGA and Windows compatible display capable of displaying 256 colors (RGB) or more
Printer / Mouse
Any device compatible with Windows
GP-IB: a GP-IB adapter by National Instruments (AT-GPIB, AT-GPIB/TNT, or PCMCIA-GPIB)
RS-232: a built-in port, or COM1 or COM2 exclusive (COM1 only for PC9801 or PC9821)
GP-IB Interface to prevent electric shock, turn the power to the instrument OFF when connecting it to the PC. Connect a GP-IB cable from the GP-IB connector on the rear panel of the WT series instrument to the corresponding connector on the PC.
All four downloadable program files are self-extracting executables which, when double-clicked, create new files and folders on your hard drive. Two of those four programs require no further installation, but the other two create an additional Setup.exe file which must be executed to complete the installation.
Turn ON the PC and start Windows. Create a new folder on your hard disk with a name of your choice, and download the free software into the folder. Double-click the file necessary for your WT series model, then click OK in the dialog box. A new folder, YOKOGAWA, will be created on drive C and files will be extracted to it. Specify any changes you wish to make to the destination folder.
When the extraction is finished, check to see if the program executable file was created (C:\YOKOGAWA\WTx000harm20e2.exe or C:\YOKOGAWA\WT2000flick\flick.exe). If so, installation is complete. If the Setup.exe file (rather than executable files) was created you will need to perform a few more steps to complete the installation. Double-click the Setup.exe file, and follow the on-screen instructions.
After the program has been fully installed, you can use Explorer or the My Computer browser to create a shortcut to the program. Locate the executable file, right-click its icon, and select Create Shortcut from the shortcut menu. Running/Exiting the Software.
Please read Software Agreement carefully before downloading.
|Release Date||Feb 18, 2003|
|File Name||wtxgne.zip (for GP-IB)||2.3 MB|
|wtxrne110.zip (for RS-232C)||2.3 MB|
Downloading of the software indicates acceptance of the software agreement.
The WT2000 digital powermeter series has been designed with emphasis on basic performance (bandwidth, accuracy, response speed, and noise immunity) from the viewpoint of measurement of electrical quantities. The broad range of functions of these power analyzers enable them to be used in various fields of applications.
Superior cost performance and compact size make the entry level WT200 Series Digital Power Meters ideal for manufacturing and basic power measurement applications. Offered with One to Three Elements, with Basic Power Accuracy of 0.1% and 100kHz Bandwidth.