You can check the format of the eSATA drive through the Windows Disk Management service. If the drive format is NTFS, you will need to format the drive to FAT32. Please backup any files on the eSATA drive to separate location. The formatting process will delete all contents of the eSATA Drive. It is recommended to format the eSATA drive using the SATA cable provided with the drive whether you are formatting using the PC or the ScopeCorder. We cannot guarantee the eSATA drive will format properly if you do so through USB.
To perform the format check in Windows, please follow the procedures listed below:
A list of compatible eSATA drives with the ScopeCorder is:
Data acquisition (DAQ) is the measurement, recording, analyzing, and presentation of real world phenomena. It includes electrical measurements such as voltage, current, power as well as measurements through sensors and transducers including temperature, pressure, strain, vibration, and more.
Yokogawa high speed data acquisition systems deliver industry leading isolation, bit resolution, sampling rate, and memory depth, with independent channel hardware and easy to use software.
Capture, analyze, and troubleshoot electrical or real-world physical signals. Oscilloscopes observe the change of electrical signals over time, continuously graphed on a display as voltage or amplitude vs. time.
A ScopeCorder is an instrument combining a mixed signal oscilloscope and portable data acquisition recorder into a modular platform designed to capture both high-speed transients and low-speed trends. Yokogawa’s ScopeCorder product family provides flexible and high-performance multi-channel test instruments by combining a variety of signal conditioning input modules, onboard calculations, and deep data acquisition storage into an off-the-shelf data acquisition solution.