# What is the Update Cycle of the D/A Output of the WE7141 100Mhz Universal Counter Module?

The update cycle changes depending on the acquisition mode and the gate time during frequency measurement. When the acquisition mode is One Shot, the update cycle is approximately 0.5-1 seconds. However for frequency measurement, it's about 10 seconds when the gate time is 10 seconds. When the method is Free Run, the update cycle occurs at each sampling interval. However for frequency measurement, it occurs approximately at each (gate time + sampling interval).

### 相关产品和解决方案

#### 数据采集设备

Yokogawa数据采集系统具备出众的灵活性和性能，能测量、显示、存储甚至促动各种物理和电气现象。

#### 高速数据采集器和数采仪

Yokogawa高速数据采集系统提供业界领先的隔离、位分辨率、采样速率和存储长度，带有独立的通道硬件和使用简单的软件。是示波记录仪中最领先的技术。