Portable and Handheld Instruments

- Digital Multimeters
- Calibrators
- Portable Power Monitor/Recorder
- Clamp-on Testers
- DC Precision Measuring
- Insulation Testers
- Earth Ground Testers
- Lux Meters
- Portable Analog Instruments
- Thermometers
- Process Calibrator CA300 Series
- Meters Accessories
- Digital Multimeters Accessories
- Calibrators Accessories

The 2406E Analog Insulation Tester:

- Analog models with two and three rating
- AC voltage measurement
- Automatic discharge
- Sky blue EL backlight
- Increased safety (covered battery charger)

Rating | 125V/20M~ 25V/5M~ 50V/10M~ |
---|---|

Effective Measuring Range | 0.001-5M~ 0.005-10M~ 0.01-20M~ |

Central Scale Value | 0.1M~ 0.2M~ 0.5M~ |

AC Voltage Measurement | 0 to 300 V |

Lower Limit | 0.025M~ 0.05M~ 0.125~ |

Rated Current | 1 mA |

Approx. Outer Scale Length | 88 mm |

Discharge Function | The tester automatically begins discharge when you turn off the MEAS switch. The pointer swings if there is any residual voltage in the circuit under test. You can make sure discharge is complete by checking that the pointer swings back to the infinite (& |

Approx. Width | 120 mm |

Approx. Height | 110 mm |

Approx. Depth | 60 mm |

Approx. Weight | 500 g |

Batteries | Six AA (R6P) batteries |

Standard Accessories | Batteries (Qty- 6) Carrying case (B9075MU) (Qty -1) Earth probe (blake); approx. 1 m long Earth probes and Line Probes (98007) (Qty - 1) User's Manual (Qty - 1)- With probe-housing pocket and neck strap |

Ambient Temperature/Humidity Ranges | 23 ±5 /45-75% RH |
---|---|

Position of Use | Horizontal (5º max. of angle of inclination) |

External Magnetic Fields | None Battery voltage: Within effective voltage range (The pointer must stay within the range indicated by the BAT symbol when the battery check is performed.) |

Tolerance Mentioned Conditions | Resistance measurement: First effective measuring range = ±5% of reading Second effective measuring range = ±10% of reading Infinite and zero scale values: 0.7% max. of scale length AC voltage: ±10% of maximum scale value No-load voltage: 130% max. of rated voltage Rated measuring current: 1 mA (0 to 20%) when in first effective measuring range Short-circuit current: 12 mA max. |

Rating | 125V/20M~ 25V/5M~ 50V/10M~ |
---|---|

Effective Measuring Range | 0.001-5M~ 0.005-10M~ 0.01-20M~ |

Central Scale Value | 0.1M~ 0.2M~ 0.5M~ |

AC Voltage Measurement | 0 to 300 V |

Lower Limit | 0.025M~ 0.05M~ 0.125~ |

Rated Current | 1 mA |

Approx. Outer Scale Length | 88 mm |

Discharge Function | The tester automatically begins discharge when you turn off the MEAS switch. The pointer swings if there is any residual voltage in the circuit under test. You can make sure discharge is complete by checking that the pointer swings back to the infinite (& |

Approx. Width | 120 mm |

Approx. Height | 110 mm |

Approx. Depth | 60 mm |

Approx. Weight | 500 g |

Batteries | Six AA (R6P) batteries |

Standard Accessories | Batteries (Qty- 6) Carrying case (B9075MU) (Qty -1) Earth probe (blake); approx. 1 m long Earth probes and Line Probes (98007) (Qty - 1) User's Manual (Qty - 1)- With probe-housing pocket and neck strap |

Ambient Temperature/Humidity Ranges | 23 ±5 /45-75% RH |
---|---|

Position of Use | Horizontal (5º max. of angle of inclination) |

External Magnetic Fields | None Battery voltage: Within effective voltage range (The pointer must stay within the range indicated by the BAT symbol when the battery check is performed.) |

Tolerance Mentioned Conditions | Resistance measurement: First effective measuring range = ±5% of reading Second effective measuring range = ±10% of reading Infinite and zero scale values: 0.7% max. of scale length AC voltage: ±10% of maximum scale value No-load voltage: 130% max. of rated voltage Rated measuring current: 1 mA (0 to 20%) when in first effective measuring range Short-circuit current: 12 mA max. |

Rating | 125V/20M~ |
---|---|

Effective Measuring Range | 0.01-20M~ |

Central Scale Value | 0.5M~ |

AC Voltage Measurement | 0 to 300 V |

Lower Limit | 0.125~ |

Rated Current | 1 mA |

Approx. Outer Scale Length | 88 mm |

Discharge Function | The tester automatically begins discharge when you turn off the MEAS switch. The pointer swings if there is any residual voltage in the circuit under test. You can make sure discharge is complete by checking that the pointer swings back to the infinite (& |

Approx. Width | 120 mm |

Approx. Height | 110 mm |

Approx. Depth | 60 mm |

Approx. Weight | 500 g |

Batteries | Six AA (R6P) batteries |

Standard Accessories | Batteries (Qty- 6) Carrying case (B9075MU) (Qty -1) Earth probe (blake); approx. 1 m long Earth probes and Line Probes (98007) (Qty - 1) User's Manual (Qty - 1)- With probe-housing pocket and neck strap |

Ambient Temperature/Humidity Ranges | 23 ±5 /45-75% RH |
---|---|

Position of Use | Horizontal (5º max. of angle of inclination) |

External Magnetic Fields | None Battery voltage: Within effective voltage range (The pointer must stay within the range indicated by the BAT symbol when the battery check is performed.) |

Tolerance Mentioned Conditions | Resistance measurement: First effective measuring range = ±5% of reading Second effective measuring range = ±10% of reading Infinite and zero scale values: 0.7% max. of scale length AC voltage: ±10% of maximum scale value No-load voltage: 130% max. of rated voltage Rated measuring current: 1 mA (0 to 20%) when in first effective measuring range Short-circuit current: 12 mA max. |

Rating | 250V/50M~ |
---|---|

Effective Measuring Range | 0.01-50M~ |

Central Scale Value | 1M~ |

AC Voltage Measurement | 0 to 300 V |

Lower Limit | 0.25~ |

Rated Current | 1 mA |

Approx. Outer Scale Length | 88 mm |

Discharge Function | |

Approx. Width | 120 mm |

Approx. Height | 110 mm |

Approx. Depth | 60 mm |

Approx. Weight | 500 g |

Batteries | Six AA (R6P) batteries |

Standard Accessories | Batteries (Qty- 6) Carrying case (B9075MU) (Qty -1) Earth probe (blake); approx. 1 m long Earth probes and Line Probes (98007) (Qty - 1) User's Manual (Qty - 1)- With probe-housing pocket and neck strap |

Ambient Temperature/Humidity Ranges | 23 ±5 /45-75% RH |
---|---|

Position of Use | Horizontal (5º max. of angle of inclination) |

External Magnetic Fields | |

Tolerance Mentioned Conditions | Resistance measurement: First effective measuring range = ±5% of reading Second effective measuring range = ±10% of reading Infinite and zero scale values: 0.7% max. of scale length AC voltage: ±10% of maximum scale value No-load voltage: 130% max. of rated voltage Rated measuring current: 1 mA (0 to 20%) when in first effective measuring range Short-circuit current: 12 mA max. |

Rating | 125V/20M~ 250V/50M~ 500V/100M~ |
---|---|

Effective Measuring Range | 0.01-20M~ 0.01-50M~ 0.05-100M~ |

Central Scale Value | 0.5M~ 1M~ 2M~ |

AC Voltage Measurement | 0 to 600 V |

Lower Limit | 0.125M~ 0.25M~ 0.5~ |

Rated Current | 1 mA |

Approx. Outer Scale Length | 88 mm |

Discharge Function | |

Approx. Width | 120 mm |

Approx. Height | 110 mm |

Approx. Depth | 60 mm |

Approx. Weight | 500 g |

Batteries | Six AA (R6P) batteries |

Standard Accessories | Batteries (Qty- 6) Carrying case (B9075MU) (Qty -1) Earth probe (blake); approx. 1 m long Earth probes and Line Probes (98007) (Qty - 1) User's Manual (Qty - 1)- With probe-housing pocket and neck strap |

Ambient Temperature/Humidity Ranges | 23 ±5 /45-75% RH |
---|---|

Position of Use | Horizontal (5º max. of angle of inclination) |

External Magnetic Fields | |

Tolerance Mentioned Conditions | Resistance measurement: First effective measuring range = ±5% of reading Second effective measuring range = ±10% of reading Infinite and zero scale values: 0.7% max. of scale length AC voltage: ±10% of maximum scale value No-load voltage: 130% max. of rated voltage Rated measuring current: 1 mA (0 to 20%) when in first effective measuring range Short-circuit current: 12 mA max. |

Rating | 125V/20M~ 250V/50M~ 500V/100M~ |
---|---|

Effective Measuring Range | 0.01-20M~ 0.01-50M~ 0.05-100M~ |

Central Scale Value | 0.5M~ 1M~ 2M~ |

AC Voltage Measurement | 0 to 600 V |

Lower Limit | 0.125M~ 0.25M~ 0.5~ |

Rated Current | 1 mA |

Approx. Outer Scale Length | 88 mm |

Discharge Function | |

Approx. Width | 120 mm |

Approx. Height | 110 mm |

Approx. Depth | 60 mm |

Approx. Weight | 500 g |

Batteries | Six AA (R6P) batteries |

Standard Accessories | Batteries (Qty- 6) Carrying case (B9075MU) (Qty -1) Earth probe (blake); approx. 1 m long Earth probes and Line Probes (98007) (Qty - 1) User's Manual (Qty - 1)- With probe-housing pocket and neck strap |

Ambient Temperature/Humidity Ranges | 23 ±5 /45-75% RH |
---|---|

Position of Use | Horizontal (5º max. of angle of inclination) |

External Magnetic Fields | |

Tolerance Mentioned Conditions | Resistance measurement: First effective measuring range = ±5% of reading Second effective measuring range = ±10% of reading Infinite and zero scale values: 0.7% max. of scale length AC voltage: ±10% of maximum scale value No-load voltage: 130% max. of rated voltage Rated measuring current: 1 mA (0 to 20%) when in first effective measuring range Short-circuit current: 12 mA max. |

Rating | 1000V/2000M~ 250V/50M~ 500V/100M~ |
---|---|

Effective Measuring Range | 0.01-50M~ 0.05-100M~ 1-2000M~ |

Central Scale Value | 1M~ 2M~ 50M~ |

AC Voltage Measurement | 0 to 600 V |

Lower Limit | 0.25M~ 0.5M~ 1~ |

Rated Current | 1 mA |

Approx. Outer Scale Length | 88 mm |

Discharge Function | |

Approx. Width | 120 mm |

Approx. Height | 110 mm |

Approx. Depth | 60 mm |

Approx. Weight | 500 g |

Batteries | Six AA (R6P) batteries |

Standard Accessories | Batteries (Qty- 6) Carrying case (B9075MU) (Qty -1) Earth probe (blake); approx. 1 m long Earth probes and Line Probes (98007) (Qty - 1) User's Manual (Qty - 1)- With probe-housing pocket and neck strap |

Ambient Temperature/Humidity Ranges | 23 ±5 /45-75% RH |
---|---|

Position of Use | Horizontal (5º max. of angle of inclination) |

External Magnetic Fields | |

Tolerance Mentioned Conditions | Resistance measurement: First effective measuring range = ±5% of reading Second effective measuring range = ±10% of reading Infinite and zero scale values: 0.7% max. of scale length AC voltage: ±10% of maximum scale value No-load voltage: 130% max. of rated voltage Rated measuring current: 1 mA (0 to 20%) when in first effective measuring range Short-circuit current: 12 mA max. |

Rating | 1000V/2000M~ 250V/50M~ 500V/100M~ |
---|---|

Effective Measuring Range | 0.01-50M~ 0.05-100M~ 1-2000M~ |

Central Scale Value | 1M~ 2M~ 50M~ |

AC Voltage Measurement | 0 to 600 V |

Lower Limit | 0.25M~ 0.5M~ 1~ |

Rated Current | 1 mA |

Approx. Outer Scale Length | 88 mm |

Discharge Function | |

Approx. Width | 120 mm |

Approx. Height | 110 mm |

Approx. Depth | 60 mm |

Approx. Weight | 500 g |

Batteries | Six AA (R6P) batteries |

Standard Accessories | Batteries (Qty- 6) Carrying case (B9075MU) (Qty -1) Earth probe (blake); approx. 1 m long Earth probes and Line Probes (98007) (Qty - 1) User's Manual (Qty - 1)- With probe-housing pocket and neck strap |

Ambient Temperature/Humidity Ranges | 23 ±5 /45-75% RH |
---|---|

Position of Use | Horizontal (5º max. of angle of inclination) |

External Magnetic Fields | |

Tolerance Mentioned Conditions | Resistance measurement: First effective measuring range = ±5% of reading Second effective measuring range = ±10% of reading Infinite and zero scale values: 0.7% max. of scale length AC voltage: ±10% of maximum scale value No-load voltage: 130% max. of rated voltage Rated measuring current: 1 mA (0 to 20%) when in first effective measuring range Short-circuit current: 12 mA max. |

Rating | 1000V/2000M~ 250V/500M~ 500V/1000M~ |
---|---|

Effective Measuring Range | 0.1-500M~ 0.5-1000M~ 1-2000M~ |

Central Scale Value | 10M~ 20M~ 50M~ |

AC Voltage Measurement | 0 to 600 V |

Lower Limit | 0.25M~ 0.5M~ 1~ |

Rated Current | 1 mA |

Approx. Outer Scale Length | 88 mm |

Discharge Function | |

Approx. Width | 120 mm |

Approx. Height | 110 mm |

Approx. Depth | 60 mm |

Approx. Weight | 500 g |

Batteries | Six AA (R6P) batteries |

Standard Accessories | Batteries (Qty- 6) Carrying case (B9075MU) (Qty -1) Earth probe (blake); approx. 1 m long Earth probes and Line Probes (98007) (Qty - 1) User's Manual (Qty - 1)- With probe-housing pocket and neck strap |

Ambient Temperature/Humidity Ranges | 23 ±5 /45-75% RH |
---|---|

Position of Use | Horizontal (5º max. of angle of inclination) |

External Magnetic Fields | |

Tolerance Mentioned Conditions | Resistance measurement: First effective measuring range = ±5% of reading Second effective measuring range = ±10% of reading Infinite and zero scale values: 0.7% max. of scale length AC voltage: ±10% of maximum scale value No-load voltage: 130% max. of rated voltage Rated measuring current: 1 mA (0 to 20%) when in first effective measuring range Short-circuit current: 12 mA max. |

Rating | 1000V/2000M~ 250V/500M~ 500V/1000M~ |
---|---|

Effective Measuring Range | 0.1-500M~ 0.5-1000M~ 1-2000M~ |

Central Scale Value | 10M~ 20M~ 50M~ |

AC Voltage Measurement | 0 to 600 V |

Lower Limit | 0.25M~ 0.5M~ 1~ |

Rated Current | 1 mA |

Approx. Outer Scale Length | 88 mm |

Discharge Function | |

Approx. Width | 120 mm |

Approx. Height | 110 mm |

Approx. Depth | 60 mm |

Approx. Weight | 500 g |

Batteries | Six AA (R6P) batteries |

Standard Accessories | Batteries (Qty- 6) Carrying case (B9075MU) (Qty -1) Earth probe (blake); approx. 1 m long Earth probes and Line Probes (98007) (Qty - 1) User's Manual (Qty - 1)- With probe-housing pocket and neck strap |

Ambient Temperature/Humidity Ranges | 23 ±5 /45-75% RH |
---|---|

Position of Use | Horizontal (5º max. of angle of inclination) |

External Magnetic Fields | |

Tolerance Mentioned Conditions | Resistance measurement: First effective measuring range = ±5% of reading Second effective measuring range = ±10% of reading Infinite and zero scale values: 0.7% max. of scale length AC voltage: ±10% of maximum scale value No-load voltage: 130% max. of rated voltage Rated measuring current: 1 mA (0 to 20%) when in first effective measuring range Short-circuit current: 12 mA max. |

Model | Rating | Effective Measuring range |
Central Scale Value |
AC Voltage Measuring range |
Lower limit of measured Ω ^{*1} |
Rated Current |

240631 |
25 V/5 MΩ
50 V/10 MΩ 125 V/20 MΩ |
0.001 to 5 MΩ
0.005 to 10 MΩ 0.01 to 20 MΩ |
0.1 MΩ 0.2 MΩ 0.5 MΩ |
0 to 300 V |
0.025 MΩ
0.05 MΩ 0.125 MΩ |
1 mA 1 mA 1 mA |

240641 | ||||||

240632 |
125 V/20 MΩ
250 V/50 MΩ |
0.01 to 20 MΩ
0.01 to 50 MΩ |
0.5 MΩ
1 MΩ |
0 to 300 V |
0.125 MΩ
0.25 MΩ |
1 mA 1 mA |

240642 | ||||||

240633 |
125 V/20 MΩ
250 V/50 MΩ 500 V/100 MΩ |
0.01 to 20 MΩ
0.01 to 50 MΩ 0.05 to 100 MΩ |
0.5 MΩ
1 MΩ 2 MΩ |
0 to 600 V |
0.125 MΩ
0.25 MΩ 0.5 MΩ |
1 mA 1 mA 1 mA |

240643 | ||||||

240634 |
250 V/50 MΩ
500 V/100 MΩ 1000 V/2000 MΩ |
0.01 to 50 MΩ
0.05 to 100 MΩ 1 to 2000 MΩ |
1 MΩ
2 MΩ 50 MΩ |
0 to 600 V |
0.25 MΩ
0.5 MΩ 1 MΩ |
1 mA 1 mA 1 mA ^{*2} |

240644 | ||||||

240635 |
250 V/500 MΩ
500 V/1000 MΩ 1000 V/2000 MΩ |
0.1 to 500 MΩ
0.5 to 1000 MΩ 1 to 2000 MΩ |
10 MΩ 20 MΩ 50 MΩ |
0 to 600 V |
0.25 MΩ
0.5 MΩ 1 MΩ |
1 mA^{*2}1 mA ^{*2}1 mA ^{*2} |

240645 |

EL-Back-Light | Non-Back-Light | ^{*1} The minimum value at which the rated voltage can be maintained;^{*2} 0.55 mA in the case of the first effective measuring range |

Top