# What is the Lowest Current that can be Measured by the DL Series Oscilloscope Current Probes?

There is no specification of the lowest current for our current probe.

However, it is possible to calculate the lowest current from the accuracy of this probe.

Models 701929/701933/701928/701932

Amplitude Accuracy: +/- ( 1.0% of reading + 1mV)

Output Voltage Rate: 0.1V/A

There is a offset error of +/- 1mV. Even if the input current is zero A, there is a possibility that the output is 1mV.

The scaled current value for a 1mV signal is 10mA. Input current less than 10mA cannot be measured.

If the current being measured is 20mA, the measured deviation will be 20mA +/- 10mA, and the accuracy is 50%.

Therefore, in most cases 20mA cannot be measured.

The lowest current that can be measured accurately is 1A. The measurement accuracy for 1A input current is +/-2% or 1A +/-20mA( 1.0% of reading + 10mA).

Model 701930

Amplitude Accuracy: +/- ( 1.0% of reading + 1mV)

Output Voltage Rate: 0.01V/A

There is a offset error of +/- 1mV. The scaled current value for a 1mV signal is 100mA.

The lowest current that can be measured correctly is 10A.

The measurement accuracy for 10A input current is +/- 2% or 10A +/-200mA( 1.0% of reading + 100mA).

Model 701931

Amplitude Accuracy: +/- ( 1.0% of reading + 5mV)

Output Voltage Rate: 0.01V/A

There is a offset error of +/- 5mV. The scaled current value for a 1mV signal is 500mA.

The lowest current that can be measured correctly is 50A.

The measurement accuracy for 50A input current is +/- 2% or 50A +/-1A( 1.0% of reading + 500mA).

### 相关产品和解决方案

#### 波形测量仪器

Yokogawa在业界首推示波记录仪的概念，将数字示波器和多通道数据记录仪的优点集于一身。横河示波记录仪作为世界一流的示波记录仪具有高速采样能力和大测量带宽的优点，可广泛应用于电子工程的设计和开发过程。