WT1000 Series: perform harmonic analasis conforming to the IEC1000-3-2 standard. This software does not have the capability to distinguish the class (based on the IEC standard) of the item being measured by the level of the higher harmonics. However, you can enter the WT1000/WT2000's voltage and current range settings, and change the display in the instrument's front panel.
WT2000 Series: perform harmonic analasis conforming to the IEC1000-3-2 standard.
Allows you to perform hours of continuous measurement, 16 cycles at a time, and make evaluations at 2.5 minute intervals. In particular, you can simultaneously measure 3 phases, and this provides even more uses for the instrument. The general functions of this software allow you to enter the voltage and current range, and other settings in the instrument's display panel.
PCA PC with a 80486DX2 CPU or later (Pentium or later recommended), and 8 MB or more of memory.
|Operating system:||Windows 98, Windows Me, Windows 2000 Professional, or Windows XP|
|Floppy disk drive:||3.5" floppy disk drive using 1.44 MB (MS-DOS compatible) type disks|
|Hard disk:||A hard disk drive with 10 MB of available free space|
|Video adapter:||A video adapter capable of displaying at 640 x 480 or higher resolution (1024 x 768, 16 bit color or higher recommended)|
|Display:||VGA and Windows compatible display capable of displaying 256 colors (RGB) or more|
|Printer:||Any device compatible with Windows|
|Mouse:||Any device compatible with Windows|
|Communications interface:||GP-IB: a GP-IB adapter by National Instruments (AT-GPIB, AT-GPIB/TNT, or PCMCIA-GPIB)
RS-232: a built-in port, or COM1 or COM2 exclusive (COM1 only for PC9801 or PC9821)
GP-IB Interface to prevent electric shock, turn the power to the instrument OFF when connecting it to the PC. Connect a GP-IB cable from the GP-IB connector on the rear panel of the WT series instrument to the corresponding connector on the PC.
Turn on your PC and start Windows.
Extract the downloaded ZIP compressed file. A new folder, same as the file name (WTx000hrm), will be created on drive C and files will be extracted to it.
If so, installation is complete.
After the program has been installed, you can use Explorer to create a shortcut to the program. Locate the executable file, right-click its icon, and select Create Shortcut from the shortcut menu.
Double-click the program executable file or its shortcut to run the software.
Please read Software Agreement carefully before downloading.
Downloading of the software indicates acceptance of the software agreement.
The WT2000 digital powermeter series has been designed with emphasis on basic performance (bandwidth, accuracy, response speed, and noise immunity) from the viewpoint of measurement of electrical quantities. The broad range of functions of these power analyzers enable them to be used in various fields of applications.